Any ammeter, including a multimeter in a current range, has a certain resistance. Most multimeters inherently measure voltage, and pass a current to be measured through a shunt resistance, measuring the voltage developed across it. The voltage drop is known as the burden voltage, specified in volts per ampere. The value can change depending on the range the meter selects, since different ranges usually use different shunt resistors.
The burden voltage can be significant in very low-voltage circuit areas. To check for its effect on accuracy and on external circuit operation the meter can be switched to different ranges; the current reading should be the same and circuit operation should not be affected if burden voltage is not a problem. If this voltage is significant it can be reduced (also reducing the inherent accuracy and precision of the measurement) by using a higher current range.
Read more about this topic: Multimeter
Famous quotes containing the word burden:
“Ovid lies here, the poet, skilled in loves gentle sport;
By his own talents he worked his undoing.
Oh, you who pass by, if ever you have loved,
Think it not a burden to wish him calm repose.”
—Ovid (Publius Ovidius Naso)